ScripHessco Reusable Hot & Cold Gel Ice Pack 5" x 10.5"
$2.59 As low as $2.05
Plastic Cap with a Pop-up Pouring Spout For Bottle Sizes 2 - 16 Oz (Flip Top)
TENS 3000 Unit - Dual Channel, 3 Modes, Analog with Timer
CBD CLINIC™ Clinical Strength Pain Stick - Level 5
This item is out of stock$80.00Please Log in for professional pricing
ScripHessco Economy Smooth Headrest Paper, 8.5"W x 225'L for Chiropractor & Exam Tables
Boar Brush with Strap - Boar Bristle Exfoliating Brush
Twist Lock Jar Pump White 8 Oz
Digital Galvanic Stimulator - High-Voltage Pulsed Galvanic Stimulation
Digital Galvanic Stim High Volt
Digital Galvanic Stimulation is most useful in acute injuries associated with major tissue trauma with bleeding or swelling. In contrast to TENS or IFC units, which apply alternating current, Galvanic Stimulators apply direct current. Direct current creates an electrical field over the treated area that, theoretically, changes blood flow and moves fluids. The positive pad behaves like ice, causing reduced circulation to the area under the pad and reduction in swelling. The negative pad behaves like heat, causing increased circulation, reportedly speeding healing.
High-voltage pulsed galvanic stimulation (HVPGS) is gaining widespread use for edema reduction, wound healing, pain relief, Carpal Tunnel Syndrome and Diabetic Foot are two major areas of use. Devices in this class are characterized by a unique twin- peak monophasic waveform with very short pulse duration (microseconds) and a therapeutic voltage greater than 100 volts. The combination of very short pulse duration and high peak current, yet low total current per second (Microcurrent) allows relatively comfortable stimulation. Furthermore, this combination provides an efficient means of exciting sensory, motor and pain-conducting nerve fibers. Perceptual discrimination of those responses is relatively easy to achieve and thus its clinical versatility. Back Ground Skin offers a great amount of resistance to the flow of electrical current. When current is passed through a circuit that contains resistors, voltage drop occurs and energy is lost. This phenomenon occurs when traditional low voltage units are used in treatment A high voltage device produces a spontaneous breakdown in skin resistance and HVPGS current passes through the skin with negligible thermal and electrochemical effects. The first high voltage stimulator was developed by Bell Telephone Laboratories in 1945. By decreasing the pulse duration and increasing the voltage, the developers noted that deep tissues could be stimulated without producing tissue damage. The first published report (I 966) described its effectiveness in wound healing of animal limbs. High-Voltage Pulsed Galvanic Stimulation
A prescription from your physician or physical therapist is required for this device.Package includes:
- 1 pair of lead wires
- 1 pack of four active electrodes and 1 ground electrode
- Industrial strength heavy duty 9V battery
- Carrying case
- Users manual
- Belt clip
Muscle soreness, Muscle stiffness, Arthritis, Body pain, Post-op pain, Post-traumatic acute pain and Chronic pain.
Warranty Terms:(Please Read):
The warranty for this product covers the device itself and its power cord (where applicable). The warranty does not cover damages caused by normal wear and tear, misuse, or abuse of the device or any of its components. Please keep in mind that Lead Wires (the wires that connect the device to the electrodes) are not covered under the manufacturer’s warranty for this device. The periodic replacement of lead wires and disposable electrodes is part of the upkeep of any electrotherapy device.
I am currently under the treatment of a physician who has prescribed this product. I will read and completely follow all directions provided by the manufacturer, as well as instructions from my physician regarding treatment times, and protocols. I understand that I will assume all responsibility for the use/misuse of this product. I will contact my physician immediately in the case of any unusual reactions caused by the use of this product. I understand that Healio Health/Healio Living is only a distributor and in no way assumes responsibility for any injury it may cause due to malfunction, misuse, inappropriate application, or other reason. By clicking “add to cart” you certify that the above statement(s) is/are true.
Power: 9V battery or AC Adaptor
Output: Single Channel. Two active outputs and one dispersive return. Where active 1 and dispesive located on the output of CH1, the active 2 located on the output of CH2.
Output Voltage Adjustable, 0-350 Volts maximum (500 ohm load)
Wave Form: Twin peaked monophasic waveform with fixed interpulse interval of 100u
Pulse Rate: 1z-120Hz (adjustable), 1Hz/step
Pulse Duration: 5 uS
Interpulse Interval: 100 us
Modes: Continuous, Alternating
1 Year Manufacturer Warranty System Includes: Stimulator, Lead Wires, 4 Self-Stick Hypoallergenic Disposable Electrodes, 9V Battery and AC Adapter, Soft Carrying Case, Instruction Booklet.